12-13-2022 06:35 PM
I want to measurement 2 channel current by Agilent 34410A,how to configure front/rear enable both.
Just search in forum only have 1 channel, any information would be appreciated.
12-13-2022 08:02 PM
34410A is a 1-channel DMM, the front/back are the same connections internally routed. You need an external multiplexer to switch the signals to DMM or get another DMM.
BTW, all DMMs are 1-channel, some DAQ-like DMMs have a multiplexer/switch matrix card inbuilt to measure multiple signals.
12-21-2022 04:24 AM
Thank for you reply.
Additional question is when use NI-USB-GPIB Card how to control 34410A from remote to local model. I tried send command as "SYST:LOCAl",doesn't work and got an error.
12-21-2022 04:44 AM
I tried send command as "SYST:LOCAl",doesn't work and got an error.
What "doesn't work" and which error did you get?
12-21-2022 05:41 AM
34410 LED display "ERROR" still keep remote model .