Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
We have used 9184 chassis in the past but I am told by my peers that this is no longer available and now we are buying 9185 chassis.
Meanwhile, I have a system which needs a voltage input and a current input module (9205 and 9208 are the modules that I have in hand).
I have been told my 9205 will not work with the newer 9185 chassis. And from what I can tell the 9208 is not recognized in the newer 9185 chassis.
So it seems like I need the old 9184 chassis for both of these modules that I have.
Can NI point me to a chassis vs. module compatibility chart?
If I have a 9184 chassis what voltage and current input modules can I use?
If I have a 9185 chassis what voltage and current input modules can I use?
I worry I am buying the wrong parts.
Go to Solution.
My colleague came in. Apparently the incompatibility is actually with software.
Once I upgraded to NIDAQ1710f0 the 9185 chassis can recognize my modules.
What do you need our team of experts to assist you with?
We'll be in touch soon!