Hi Every one
I'm new to labview.
I use a high accuracy load cell and signal conditioner which provide precise excitation and output (10 V)
The problem that when I use the DAQ card (myDAQ) to measure the output voltage
The voltage reaches zero for finite time as in the image however it is very stable at zero (at no load)
Also the same problem I found earlier when measuring a 9V DC battery
Solved! Go to Solution.
Please post a diagram showing exactly how everything is connected. Also pst yoru VI with some typical data saved as default.
my signal conditioner is programmed via serial port
when I connect it to pc via usb-232 converter in the same time of aqcuiring data then all readings are ok and no ripples or spikes to zero
The connection diagram and VI block are attatched
I suspect that you have a grounding problem. The signal conditioner is probably grounded through the USB/RS-232 connection when you have that connected. When it is not connected through the USB, it may not be grounded.
Try connecting the AGND of the MyDAQ to the 24 V DC- connection or to a ground connection on the signal conditioner.
Thank you for letting us know. When you report what solved the problem for you, it can help someone else with a similar problem.