Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I got a Pickering LXI 65-200-002 chassis, connected to the PXI controller via tcp/ip. It shows up under Network Devices in MAX.
It works on the VISA Test Panel. Return the correct ID.
I want to use it in the TestStand.
When I tried to create a virtual device under NI Switch Executive, I got an error 0xBFFA400C.
I've already created IVI Driver Sessions/Logical Name for the Pickering LXI chassis.
I got other Pickering PXI boards on the same PXI chassis. All works fine.
I can create virtual device for NI Switch devices without problem.
Anyone has the same problem before?
Go to Solution.
NI MAX 21.5
Switch Executive 2020
Pickering pi40iv64.dll 4.0.18
Problem solved with the help of Pickering.
Ran Pickering diagnostic tool, the report indicates that there were old version of pi40iv remain.
Uninstall pi40iv, and manual search & delete all pi40iv dll/lib.
Ran diagnostic tool again, make sure no old version remain.
What do you need our team of experts to assist you with?
We'll be in touch soon!