Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Ive made a program that collects the information in TEDS sensors with an NI RIO 9233, the only problem is that when I change to another sensor or the direction of the sensor
I dont get any new results. Then I need to restart in the Measurment & Automation explorer.
So my question is if there are any function directly in labVIEW where I can either restart or refresh the configuration of the NI USB device?
I am a little bit new with labview so take it slow 😉
To programatically reboot your cRIO:
Reset DAQ device
Hope this helps.
What do you need our team of experts to assist you with?
We'll be in touch soon!