Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
The issue is the XY graph that is in the 'program temcont.vi'. Delete it and create a new XY graph using the output of the 'Messung temp.vi'.
It worked! Thanks! 🙂
Glad to hear it! My guess is that something similar to this post happened, maybe someone else can find the root cause.
The cause was the difference in display format between the XY graphs. The problematic XY graph was set to decimal rather than floating point:
XY graph in 'program temcont.vi'XY graph in 'Messung temp.vi'
The problem is described in more detail here. Basically, the output format inherits from the axis formatting and I have complained about it in the past elsewhere.
What do you need our team of experts to assist you with?
We'll be in touch soon!