Virus scan in progress. Please wait to download attachments.
This example works to set all DSA devices and their available channels to the first non-zero excitation level.
This code will set all DSA devices and their available channels to the first non-zero excitation level. This can be set to run as a start-up program. The goal is to run this before acquisition while the sensors are connected to allow for sufficient settling time and to allow the sensor to reach steady state.The necessary EXE file is attached, along with the source code VIs.
LabVIEW 2012 or compatible
NI-DAQmx 9.5.5 or compatible
NI Multifunction DAQ Device
Steps to Implement or Execute Code
Configure the parameter for the right channel and device.
Connect your hardware appropriately.
Run the VI.
Additional Information or References
**This document has been updated to meet the current required format for the NI Code Exchange.**
Adam Academic Product Manager National Intruments
Example code from the Example Code Exchange in the NI Community is licensed with the MIT license.