I am having trouble in getting the measurement as expected. Please go through the further to get a detailed idea.
Measurement Trigger - Measurement should start at the falling edge (which happens first)
Start Trigger - Sequence should be produced at the rising edge (which happens second) of the same signal
Since I cannot configure same Trigger input (I tried) for Source and Measurement, I am Exporting my Start Trigger to one of the PXI Trig channel (/SMU/PXI_Trig0).
Sourcing happens as expected (verified using Scope) but though I get Measurement trigger (falling edge) prior to Sourcing Trigger (rising Edge), Measurement is triggered at only rising edge (which is the Start Trigger edge).
I understood that if Start Trigger is configured both Source & Measure will happen upon the Start Trigger even though we configure Measurement Trigger separately.
When I tried to use Source Trigger instead of Start Trigger my sequencing is not produced properly. Only one voltage step is produced.