Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
RE: USB-6002 & LabVIEW throughput
I want to sample 5 ADC channels, each at 4096 Hz
(5x4096 = 20,408 samples/sec) on a USB-6002 for about 10 consecutive seconds (204, 080 total samples) in LabVIEW.
given the limitations of serial USB, small 2047 FIFO size and LabView/Windows 10?
2.) The what is the definition of “Bandwidth 300 KHz’ in the USB-6002 data sheet?
Thank you !
That same spec sheet also includes an aggregate maximum sample rate of 50 Ks/s (thousands of samples per second). Your desire to do 20 KS/s is well below the maximum.
Here is document that explains NI specifications:
Specifications Explained: NI Multifunction I/O (MIO) DAQ
What do you need our team of experts to assist you with?
We'll be in touch soon!