You probably just need to read more often, or read more samples at a time. So, if you have a wait in your loop, decrease the wait time. Or increase your "Number of Samples" input to the DAQ Assistant (or perhaps, both).
Message Edited by Alan [DE] on 03-06-2006 10:36 AM
I increased the input buffer and did not get the error message any more.
I also save the data into a file and display the waveform in a wavegraph in real time. I saw there are some gaps with no points there in the wavegraph. It apears periodically. I am wondering wether there are some data that failed to be read from the DAQ board or just because the wavegraph failed to display the data?