07-17-2014 06:48 AM
I am intending to use a PCI 6040 E connected to a BNC 2010 to measure 3 different voltages as a input. I have installed the NI-DAQ mx 9.7.5 on windoes 7 (64 bit). My question is why I can not see my devices under the "Devices and Interfaces"? even by Labview, DAQ assistant, Aquire signal, anaolge input ... I can not configure the devices by this error : No supported devices found.
Any idea? Thanks
Solved! Go to Solution.
07-17-2014 09:08 AM
I have just imroved the NI max by 9.9, But still I can not see my hardware in "devices and interfeces".
07-17-2014 10:30 AM
I'v just found that the PCI card has not been installed correctly!! Now I can see the PCI's name (PCI-MIO-16E-4 ) in NI max, under the "devices and interfaces". But no information by right click, and in Labview I can not configure it again 😞
Do you think I need to install the software again?
07-17-2014 10:38 AM
You are going the wrong way. The last version of DAQmx that supports the PCI-6040 is 9.5.1. See this paper for more info: NI-DAQmx and NI-DAQ Driver Support: PCI/PXI Devices. And LabVIEW 2011 is the last version of LabVIEW that supports DAQmx 9.5.1.
07-18-2014 05:41 AM
Thank you, It is working now..