in my current project I want to generate a voltage between 0V and 10V in 3000 steps. I also want to measure 3000 voltage data (These two processes need to be synchronized and take 100ms in total). I found NI USB 6003 model product. I wonder if this product can meet what I want? Can it generate and measure 3000 voltage data in 100ms? If not, I welcome your suggestions.
Unfortunately no, because your requirement of 3000 steps in 100ms requires 30kS/s but the AO is capable of only 5kS/s
thanks for your answer. Do you know if NI has a Compact DAQ module that meets my needs? For example, I found NI USB 6211 and this module has a maximum update rate of 250 kS/s. I believe it must be suitable for my application, right?
Of course, there are a lot of other modules (incl. C-Series) that can satisfy your needs. Yes, 6211 would also satisfy your needs.