Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
just wanted to ask you a question to understand where the problem is.
Did you already figured out if the problem is in the communication between the host and the FPGA target (running in LabVIEW) or if it's a problem generated calling the DLL in the C program?
Do you get some error code when the crash occurs?
It's the same reply I've got from NI. I didn't try because now I'm far from the system where I get the error. I'll try it soon.
However I haven't crashes when calling the functions. I can call them thousands times, until the system handles are over. Then all the system begin to be unstable (especially nidaq-mx).
It's very strange that nobody in the globe hadn't this problem before. 🙂
What do you need our team of experts to assist you with?
We'll be in touch soon!