Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Hi. Can I ask how do you make the tick count reset when you start the program .Thanks in advance!
You can't. It is based on the PC clock, It starts at zero when the PC starts.
You can take the reading at the beginning of your VI and subtract it from later tick counts.
Either take a tick count at the time of program start, then another at some stage and subtract to find the number of ticks between the two events, or use the 'elapsed time' vi.
What do you need our team of experts to assist you with?
We'll be in touch soon!