What frequency do you obtain? it could be a noise.
If you connect a referenced signal, do you have these values?
Message Edité par ManuelR le 07-06-2006 04:49 AM
Did you tried with Measurement and Automation Explorer?
How do you connect the anemometer to your FP? differential mode? RSE? NRSE?
You can launch MAX from start\programs\National Instruments\Measurement & automation.
Concerning source signal wiring, you can have a look to the following link : http://zone.ni.com/devzone/conceptd.nsf/webmain/01F147E156A1BE15862568650057DF15?opendocument
Hope this helps,
Is your problem solved?
Do you need more help?