Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Does Somebody already experienced this problem?
I just want to add test a DBL measurement to SL through a LabVIEW program:
but in SystemLink, there's something weird, the value is changed.
the problem is coming from the previous VI: Add Test Step Measurement (DBL) from the SystemLink API.
If you actually look at the numeric constant with the 17 digits of precision, the value of 2.4 becomes:
Hi, ok yes I figured out that, but I wrote 2.4, it should be 2.4000000000000000000000000000000000000000000000000000000
What do you need our team of experts to assist you with?
We'll be in touch soon!