Hello, is there possible way to connect Piezotron Coupler (cable socket should be an EMC-protected type) to screw terminal NI 9232.
I will post some photos for references. Thank you.
There are some ways to do it... 🙂
Depending what you want to measure... assuming you want the rms output, you need to provide an extra powersupply (22V-30V)
But the question is, why you want to use the coupler?
The 9232 supports IEPE/Piezotron/CCLD/... sensors directly, if you want to measure the sensor output.
You can do the rms measurement in software. Just short adapter screw to BNC ... done 🙂