I'm measuring a signal which has to be followed all the time (with minimal gaps). The module ni-9251 I'm using for that purpose is 24 bit (and excellent S/N).
I'm measuring 1000 points at one shot (burst) and then, when I repeat the burst there is a long gap (~80 ms) between the end of the first burst and beginning of the second one. I have been told by NI support that the module has no memory buffer so the data is in my computer memory, so it shouldn't be such a delay.
Is anybody aware how to lower this time gap (ideally under 10ms)?
Solved! Go to Solution.
The chassis is cDAQ 9171.
To clarify a bit, I'm trying to get as much as possible to continuous measurement.
FYI, since you don't have determinism, this might be the wrong board and you might not get as many or as fast of a response as you want. I think this is the right forum for you:
Have you tried this example? You'll have to adjust the max/min voltage to ±3
C:\Program Files (x86)\National Instruments\LabVIEW 20xx\examples\DAQmx\Analog Input\Voltage - Continuous Input.vi
Thank you nanocyte, I worked fine with NI/measurement studio example: NationalInstruments.Examples.ContAcqVoltageSamples_IntClk
I still don't understand the thing about determinism, but now I have under 1ms (that's the timer precision) gaps (between bursts) in continuous voltage sampling.
Does that mean you solved your problem? If so, you might want to mark the solution as solved. If not, you might be interested in the measurement studio forum: http://forums.ni.com/t5/Measurement-Studio-for-NET/bd-p/232