I am using PXI 1065 chassis for one of my test system. I have PXI5105 scope module set in slot no:2 and configured the scope to export a signal to PXI star trigger. I get an error saying " This slot number does not support exporting star trigger for this module" (This kind of error).
I have 2 questions.
Your help is greatly appreciated !!!
In PXI Express the PXI Star signals are on the System Timing Slot, which is slot 14 of the PXIe-1065 chassis; slot 2 is a normal PXI peripheral slot. The chassis user manual has a diagram showing the location of the various slots on the PXIe-1065 chassis.
In the previous generation PXI chassis, like the PXI-1045, slot 2 was the star trigger slot.
Other ways to synchronize the module without moving it from slot 2 would probably need to involve PXI_CLK10 or the PXI Trigger bus signals.