Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
In the NI-622x specs the sampling rate of the analog inputs is specified as
250 kS/s single channel,250 kS/s multi-channel (aggregate).
That means I have 250kS/s for ALL my analog inputs together right?
So if I would use 10 analog channels I would get a sampling rate of 25kS/s per channel.
Is that correct? Thanks for your help.
You are correct, on multiplexed boards such as the 622x the sample rate is an aggregate spec meaning that it is divided across all of the channels you are using.
We also have simultaneous boards, which have a S/s/ch (samples per second per channel) spec. These boards tend to be more expensive since they have a dedicated ADC for each channel.
What do you need our team of experts to assist you with?
We'll be in touch soon!