This is a new, Windows 10, 64 bits PC. PXI-4065 is passed the self-test, but NI-DMM soft Front panel failed to launch. Everything else works fine. Any suggestion or commands are appreciated.
BTW, Chassis and NI PXIs worked fine on Windows 7, 32 bits.
Looking at this error, a common cause is having an incorrect Instrument Descriptor. Can you verify the name in IVI resources?
A way to modify the logical name of the the instrument is resting the NI MAX Configuration Database which will force MAX to look for connected devices in that moment.
Does resetting the database allow you to launch the Soft Front Panel successfully?
Thank you for the fast reply and instruction.
The NI-DMM soft front panel is launched successfully after I reset the NI MAX data following instruction from the link above.
Thanks a lot.
Glad this information was helpful to your solution! If you can mark the post above as a solution, it will help others down the road.
Have a great day!