I have PXI-1036DC + PXI-8101RT + PXI-6682 + PXI-6653 + PXI-4462 + PXI-6143. I try to install NI-sync 3.3.5 and Multifunction DAQ & DSA 9.7.0 on 8101RT and obtain the message, that this components requires MIG String 2.3.0.
What is the MIG String and where can I take it?
Solved! Go to Solution.
Which version of LabVIEW are you using? I noticed that NI-SYNC 3.3.5 is supported for LabVIEW versions 2009 and above.
Try to navigate to C:\Program Files (x86)\National Instruments\RT Images\NI-DAQmx, open the file nimigString.cdf with a text editor. If the version is 2.2.1 then modify it to 2.3.0.
If you are not able to find the previous file, try to install DAQmx first.
Thank you wery much!
My version is 12.0.1f2 32bit. I do this modification and software installed fine.