I am using TestStand 3.5 with a PXI-4070 DMM and I want to be able to use the Offset Compensated Ohms function from the TestStand IVI-C DMM steps. I cannot find this functionality from the TestStand step, even though it is shown on the front panel of the Soft Front Panel DMM, and is stated as a IVI function. Anybody with any ideas would be greatly appreciated.
The Offset Compensated Ohms function that you are looking for is more
specialized than what the IVI driver has to offer. Instead of
using IVI DMM, try NI-DMM. This is a free download off our
website, which can be found here:
NI-DMM version 2.5 for Windows 2000/XP--LabVIEW Real-Time