Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I am trying to read an analog signal from a sensor using the myDAQ. I could only manage to pick up a very faint signal. I then plug it to a signal generator, the 5V signal only reads as 88mV on the myDAQ. Has the DAQ gone bad? Thanks for any help.
Go to Solution.
As a test could you please connect the AI terminal to the +5V terminal and see the result (and nothing elce)?
Otherwise, I recommend you open an RMA request.
What do you need our team of experts to assist you with?
We'll be in touch soon!