I have a SCXI chassis with a 1200 in slot 1 ( next to power supply ) a 1100 in slot 2 and an 1162 in slot 3. Labview 5.1 and NIDaq 6.6.0 work with the 1200 and 1162 just fine. However If I connect +2volts to channel 0 on the 1100 and zero volts on channel 1 on the 1100 both NiDaq ( test ) and Labview shows +2 volts for all 32 channels.
I have tried putting the modules in different orders and checked the jumpers ( all default except analog guard ) In NiDaq config I have the channels incremented.
Does anyone have a clue what I can do to scan all 32 analog inputs? Is the 1100 board dead?