Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I am using PXle-6358 daq.
This daq is installed in PXle-1073 and two SCB-68A are used for simultaneous multi-channel measurement.
I try to collect data by receiving a trigger signal on this daq.
I am using the DAQ-USB-6251, which is used by connecting the trigger input signal to APFI 0 (pin 29) and the GND signal to AI GND (pin 30).
I wonder which pin on the SCB-68A should connect the PXle-6358 daq trigger input signal and where to connect the GND signal.
As first step, read following link. I think "Direct Feedthrough Mode" is the best setting for you.
SCB-68 and SCB-68A : DAQ Multifunction I/O Accessory Guide
"All 68 signals from the device break out directly to the screw terminals."
As a result, pin29 on PXIe-6358 is connected to pin29 on SCB68A.
What do you need our team of experts to assist you with?
We'll be in touch soon!