10-31-2013 10:19 AM
I have been getting conflicting info RE: the USB-6008 being either able or unable to support PWM. Can anyone give a definitive answer? I was intending to use the Signal Generator function to give a digital pule out from my DAQ to control the brightness of an LED. If this can't be done i would apprecate your feedback as it will mean bining a project and starting fresh.
10-31-2013 11:22 AM
The USB-6008 uses software timed digital outputs. That means that the maximum update rate is ~150 Hz. Except for heater controls with time constants of seconds, the USB-6008 is not suitable for PWM.
10-31-2013 11:23 AM