Hi, I'm using an NI 9215 module to measure the differential voltage from a 9V battery. Three of the four channels show random downward spikes, only the last ch3 signal is within spec.
Different slots and two different 9174 chassis were used, same results as shown were produced.
The module was just calibrated by NI.
Solved! Go to Solution.
My immediate guess is a grounding issue. Did you put in a bias resistor to COM on your negative channels? The BNC version has this built in.