I have a cRIO 9076 with two NI 9265 modules, one NI 9207 module, and one NI 9482 module. The issue I am having is with the NI 9265 modules. I am using an external power supply to provide 24 volts to the module but each of the output channels reads either 7 volts or 13.2 volts. I need the modules to measure 24 volts for each channel in order for the system to function properly. Does anyone know what is wrong or what I need to change to fix this problem? Thanks for your help.
What load are you connecting to? Do you have a schematic of your setup? The 9265 does have a compliance voltage of 12V, so you shouldn't get much above that.