Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I just want to know if I can use analog inout module NI 9205 for thermistor measurement. As NI9205 has no excitation current, how to carry out the measurement?
You need an external voltage source and a stable, known resistor of about the same value in the range of the thermistor for each thermistor ....
Followed by a little math ... you calculate the thermistor resistance values by measuring the voltatages of each voltage devider build by the known resistor and the thermistor.
Add another channel to read the supplied voltage U_s
Configure the channel read list like U_s,AI(x),U_s,AI(x+1),U_s,Ai..... if you supply isn't stable 😉
Since usual thermistors aren't high precision anyway .. I guess 1% resistors should be fine...
What do you need our team of experts to assist you with?
We'll be in touch soon!