Hello: I use PCI-6024E board to do some work.when I use function "scan_op" of multichannels sample,I find I can't get a value in the help files.One of parameters of function "scan_op"---"scanRate" is so described:"When scanRate is not 0, scanRate must allow a minimum delay between the last channel of the scan and the first channel of the next scan." I want to ask how much is the value of minimum delay of PCI-6024E?If you know,please tell me,thanks.
For E Series devices, this delay corresponds to the speed of the board. So, for the 6024E case, as the maximum Sample Rate is 200kS/s, that means that the minimun delay will be 1/200k = 5u seconds. Hope this helps. Filipe Applications Engineer National Instruments