My NI DAQ 4462 current temperature is showing 0 degree when seeing through NI-MAX software.
Other Slots temperatures are showing correctly.
Can anyone tell what could be the issue ?
Does the self-calibrate, self-test and reset features work properly?
I have seen this on instruments that are messed up beyond recovery and need to be sent to NI for repair. This just means that the instrument could not read temperature and hence the default 0 value.