I was only able to get the correct signal reading by AM#!# addressing scheme while using AMUX-64T. The "O:1" addressing skeme gave ghost signal. But I do have to read mutliple channels. What could I possibly do? I am using LabView6i and NI-DAQ 6.9.1.The DAQ board is PCI-6034E.
When you use the 0:1 addressing scheme, it will read read the first 8 channels from the AMUX-64T. It will store the data in a 2D array with 8 columns. If some of those channels are not connected, the signals will be floating. If you only need 2 of these channels, you would extract them from the array. This is the only way that I know of to read multiple channels with an AMUX-64T.