If anyone has experience using these instruments or has experience with creating tracing Vi's please help Petmar.
Thanks for showing concern. I'm hoping some of the expects here will help me out.
In the Mercury_GCS_Sample_Application_1, inside the case structure, you can set the MOV subVI to move a specific amount of mm. Then you should be able to set a wait time in the General Wait subVI. Then you would sync the timing with the Mercury_GCS_Sample_Application_2a VI.
Thank you very much. I'll look into it and get back to you.
I got help from PI's resident LabVIEW expect yesterday. I'm now fixing a few bugs in the VI and will contact you when I need help. Thanks so much for your help.
Hi, I'm also trying to control a PI device with LabVIEW, could please let me know how did you make it work? I'm very confused right now cause I can't find the driver for this manufacturer.