I was able to find the IMAQdx Unconfigure Acquisition.vi. Below is a screenshot of the palette showing its location:
It basically removes any configuration information from the IMAQdx session. To best understand the configuration information it removes, it is helpful to look at the context help for IMAQdx Configure Acquisition.vi:
We can see here how the Continuous? and Number of Buffers parameters are used to configure the session. Continuous? will specify whether or not the camera continually acquires images at a hardware timed rate, and Number of Buffers will specify memory allocation for the session. The IMAQdx Unconfigure Acquisition.vi will clear out these parameters from the session.
Message Edited by Luke H on 03-12-2007 02:29 PM
I run a VI, I receive Errors, saying that ... '' open camera , close camera , get iamge, etc.'' all this functions are missing and the Sub VI s are not executable.
I dont know how to solve these errors :'( ... could any body help me?
It looks like you are missing the Imaqdx drivers. First, check that they are installed by going to Measurement and automation explorer and expanding the software tab. Then, look for this:
If it is installed, go to NI License Manager and check for the vision acquisition software--make sure it is activated.
If it is installed and activated, try repairing the install. It is possible the files were not put in the correct locations, which is why you are seeing that problem.
If you do need to install it, you can locate it here: http://joule.ni.com/nidu/cds/view/p/id/2137/lang/en
Thank you very much D Smith.
The problem is that Vision acquisition software is not activated , I just realizes that the Academic campus version that I have does not have a license for that , so I ordered the vision acquisition software, with 75 % discount :womanvery-happy: , for 102.25.
Well let me see if my program will run after activating that.