Hi, I'm new to NI and LabVIEW
I got a cDAQ-9185 with 4 modules:
- NI 9201
- NI 9219
- NI 9411
- NI 9472
I'm troubling with wiring and setting up a linear potentiometer in LabVIEW which goes from 0-8k ohm with 3 wires.
Go to Solution.
This is actually an instrumentation question and not a LabVIEW question. You don't wire a potentiometer to LabVIEW.
One side to analog ground.
Other side to +10V DC power source. (Assuming DAQ card handles 0-10 or +/- 10V DC range.)
The wiper to the analog input of your DAQ card.