I am trying to calibrate a NI 9211 unit but I keep getting the following error code.
Error or termination occurred. This calibration may not be valid. Error code: -200081 Error message: Error -200081 occurred at DAQmx Start Task.vi at step Verify Analog Input:1 Number of Channels: 2 Sample Rate: 7.140001 Maximum Sample Rate: 6.000000
I am using CalExec version 3.3.2. Is there a fix to this problem?
Ryan A. Saunders
Idaho National Laboratory
Standards and Calibration Lab
Which DAQ Chassis are you using??
Are you are using a single Module Carrier such as a NI USB-9162 or are you using a cDAQ chassis such as the NI cDAQ-9172?
Calibration is only possible will a Chassis such as the 9172 as this is not available for the single module carrier such as the 9162.
If you are able to, try using a Chassis to perform the calibration instead.