Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I am working on a project in which I am going to use thorlabs QPD (PDQ80A) to
track the position of a colloidal particle in an optical tweezer. I am using a ±12V
external power supply (Thorlabs, LDS12B) to power up the QPD. I am having a
hard time in connecting the QPD to NI SCB-68A board.
Does anyone here had the experience in hooking up this QPD
to the NI board I mentioned and read out the data using LabView?
This is the QPD that I am using https://www.thorlabs.de/thorproduct.cfm?partnumber=PDQ80A
and this is the NI board I am using https://www.ni.com/de-de/support/documentation/supplemental/17/scb-68-and-scb-68a---daq-multifunctio...
I don't have experience with thorlabs QDP, but I have experience with NI DAQ SCB 68 products.
From What i was able to grasp from the manual, the pinout gives you the signals you need to connect the two :
Pin 1 , 2 - -4 to +4V
pin 3 - 0 to 4 V
pin 4 - +5 V
pin 5 - Common
pin 6 - -5V
pins 1 , ,2 , 3 you connect to the Analog input of the NI DAQ.
The setup has to be differential to read the negative values .
Check page 2-6 of the SCB68 manual for description of the connections.
Without programming, you can use the NI max to quick test the signals coming from the QPD for sanity check before jumping into labview programming.
What do you need our team of experts to assist you with?
We'll be in touch soon!