Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I am trying to setup a NI 9216 RTD module in a cDAQ chassis. DAQmx Create Virtual Channel allows me to set the RTD type as 2-wire, 3-wire or 4-wire:
The Getting Started Guide says that the NI 9216 auto-detects whether 3- or 4-wire RTDs are connected but there does not seem to be an "auto-detect" option in the Create Virtual Channels vi.
If I leave the resistance configuration input unwired I get an error message
"Requested value is not a supported value for this property. The property value may be invalid because it conflicts with another property."
I have tried un-wiring some of the VI inputs such as excitation source but I then get a message saying the default input is incompatible with the 9216, so I cannot do without them.
I get the supported value message regardless of whether or not an RTD is physically connected to the inputs.
Am I missing something?
You are correct the 9216 will auto detect.
However, you still need to configure the DAQmx Create Virtual Channel (VI). The default within LabVIEW will be 2-wire as this is the first value offered in the enum connected to this input.
Not sure how you would get LabVIEW to do detect if it's a 3 or 4 wire configuration. There will be a way of doing this but you will probably need to use DAQmx property nodes.
I'm using the NI-9217 which has a similar function described in the data sheet.
Did you find a solution to configure the detection of the Wiring in Labview?
What do you need our team of experts to assist you with?
We'll be in touch soon!