I use the PCB 356A63 accelerometer with the Ni9231 Module, but i get always the this Daq mistake code 200284 and i increased the sample rate and decreased the the read timming rate but the mistake i still coming , can anyone help please.
There is no such Error Code. Might you mean -200284, an error code that suggests your LabVIEW code (which, unfortunately, I cannot "see" or try to execute as you failed to attach it) is "broken".
Attach your VI (not a picture of it, but the VI, itself). Attach enough "context" that we can understand where and how it is used (if it is part of a small LabVIEW Project, compress the folder holding the Project and attach the resulting .zip file).
i use the ni 9231 modul with an external powersupply for my accelerometer, when i try to measure it on NI-Max Daq and choose external power supply it gives ein error "200077".
The second thing i have signal noise on the modul, if i use the ni max daq and measure the acceleration i have a signal noise, even when i disconnect the accelerometer i still have a signal noise.
Does anyone have an idea please.
A schematic of your connection and the type of external amplifier would be helpful
Why the setting in the DAQ Task is set to internal?
you already got a comment on your noise problem in the German LabVIEW forum...
What happens when you short the unused/floating input channel?
PCB Piezotronics manufactures precision accelerometers to measure vibration, shock, acceleration, and motion for monitoring, control, and testing applications. ... Accelerometers are used by field test, in factory production and design as well as Research and Development organizations to acquire critical data.