Does anyone have experience communicating with a Newtons4th PSM1700 PsimetriQ frequency analyzer? I am trying to set a neutral bias value of 1.25VDC, but I have struck out so far.
I am interfacing with LabVIEW 2015 via a GPIB bus.
Thanks in advance for any help.
Are you using VISA to communicate to the hardware? Could you post your VI?