I have LB 2013 and Measurement and automation 18 and some hardware as NI usb 9162, NI 9237, NI 9233, NI 6009, etc...
Recently I both the NI cdaq 9174 and the temperature module NI 9210
The NIcdaq 9174 was recognised in the system, but the NI 9210 no.
As this new hardware came with new NI daq mx 19.0 I started install, but after some time it ask me if I want to replace the older version but it advise maybe this new version does not support some old harqware.
Can some one help to choose the best way to put the Ni 9210 working without loosing the support in the older harware?
The NI-9210 Spring Terminal Variant is only supported by NI-DAQmx 18.5 and later. Upgrade your version of DAQmx through our website.
You can find its supported operating systems and environments in the NI-DAQmx 18.5 readme.
More information: cDAQ Module Shows Up as Unsupported Module in NI MAX
Thanks for the help.
Seems the NI-DAQmx does not support my NI 9233, or I am wrong?
After installing NI DAQmx 19.0, now the NI9210 is recognised.
The NI 9233 is only recognised with NI USB 9162 but not with the new NI cDAQ-9174
I still need help in this subject.
After watching this https://www.ni.com/product-documentation/8136/en/ I can see that the new module Ni 9210 it is not compatible with chassi Ni 9162....