Hi! I want to use labview to read voltage from my Agilent 34450A multimeter. However, when I roll down the "Visa Resource Name" at the front panel, it does not read my multimeter, just "COM1". I know thought that the computer does read the multimeter. The Measurement & Automation Explorer does not recognize it, if that is what you are talking about. I am pretty sure the multimeter works fine, as I was able to read data through Agilent software. What can I do
Go to MAX>>Tools>>VIS>>VISA Options
Select General settings Passports. enable tulip support.
Look at the "Conflict manager page" Select NI VIAS as the prefered VISA for the interface the Agilient shows up in.
(All of this is contained in the Agilient I/O Libraries documentation under "Using with NI" or some heading like that
It should be a USBTMC device not a com port
Do you have IVI and VISA installed?
If so, when you plug in the device windows will do the driver dance and you should end up with a "device installed successfully" balloon...
then in MAX, drop down "Devices and Interfaces" and look for your instrument there:
The VISA string is that long "USB0::" by default, but you can change that to something shorter/more recognizeable.
Drop down the "Software" tab in MAX and make sure you've got latest versions of IVI and VISA, then try it again. Might need to go into Device Manager and reinstall the driver.
OK Lets clarify some things.
That device is capable of USB, RS-232 and GPIB. What interface is enabled? You will need to go to the utility menu on the front panel.
OK you are connect to a USB port on your computer, What are you connected to on the device? There are three communication interfaces available and each has the capability of being connected to a USB port through eithe a straght USB A-USB B cable or an appropiate adaptor. Knowing this would let you find it in MAX since it may be exposed in three differernt places.
You said Agilient software communicates with the device. Which agilient software. I can think of four packages that would communicate with the instrument, (I/O Libraries config manager, Connection expert, Intuilink and the new one for DMMs)