Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
I'm confused about the functionality of the niDC Power Configure Voltage Level Range VI -- For 4140 SMUs an input value x specifies a voltage range from -x to +x correct?
I'd like to range from 0 to +x only (disabling the output of negative voltages). Is there an ni power configuration I can use to achieve this?
The Configure Voltage Level Range VI is not really meant to set an allowed output range. Instead it helps the device determine what the maximum output it should be prepared to output.
You can read more about ranges here:
If you do not want to output negative voltages, the simplest way to do this is to avoid calling Configure Voltage Level.vi with a negative voltage level input parameter.
What do you need our team of experts to assist you with?
We'll be in touch soon!