I'm getting errors on start Task vi see attach DAQmx 1
I'm usingLabVIEW 14 and a USB 6009 and 2 USB-TC01 all 3 are in MAX
the program is intended to address 2 16 channel Mux chips and display 30 + channels of data. VI attached too
I have tried many hours to solve the issue of no task name, somtimes it works but then the errors return. I hope to get some help point out what I'm doing wrong
Thanks in advance
Solved! Go to Solution.
You can only have 1 analog input task on a DAQ. But you can have multiple channels in a task. For instance, for the Dev3 channels, have the channel be "Dev3/ai0..2". That will have all 3 of your channels in a single task. Then you just read all of them in a single DAQmx Read (N Channels, 1 Sample).
Thanks for the rapid responce! I'll do some rearranging and try your solution. I'll be sure to post the results
Crossrulz, your information was the solution to my problem thanks!. I had to rearrange the array scheme, terminate the tasks, handle errors etc but I think this is going to work