While running my RT single frame program to read ENI file I am getting error 4330, details of error is given below. As part of my hardware i have NI 9144 chassis to which NI 9205 and NI 9215 are slotted. Currently i just want to read my input using slaves and obtain bit information. Could you please check the error and let me know route cause and solution for the same?
4330 error :
Error - task 'AnalyzeSystem' is scheduled and could not be disabled. This task will decrease system performance. Please do either
- start uploader with administrator rights
- call schtasks.exe /Query /TN "\Microsoft\Windows\Power Efficiency Diagnostics\AnalyzeSystem" /DISABLE
<b>Complete call chain:</b>
Please attach the full VI instead of a picture of part of the VI. We cannot "click" on a picture to see the relevant Help file, we cannot execute the picture, and we cannot see the "missing stuff" that is off the edge of the picture. We also cannot tell what version of LabVIEW you are using (but the VI would tell us).
Can you explain a bit more what you are trying to do? There are LabVIEW tools for sending data from the Remote to the Host, such as Network Streams, that work extremely well ...
Please find Attached VI. Currently i just want to read my analog slaves connected. I am not sure on why the error is generated.
Please let me know if i need to give any other information.
thank you for the VI.
It seems that the Ackermann Automation lib is not a free downloadable lib. I could not find it in the internet to check your VI completly.
As far asI see you are trying to create an EtherCAT Master, don't you?
The NI 9144 is a Slave only device and can not configured as a Master.
Maybe that couses the error in the Ackermann VIs.
When me and my team analysed the error we got to know that we had problem with network adapter. So we changed adapter and set IP address to static in the network controller. Which resolved our issue.