I am communicating to the device using RS232 at 57600K.The program makes contact with the agilent device
as the front panel of the device shows RMT or something similiar which shows its in remote mode.There are no error codes.
This morning tried one of the switch programs from the drivers and it did open or close the channel switch 101 when the led was lit or unlit.
But not yet managed to get any voltages up on the screen that the channels are connected to,channels 101 and 102 for example.
The NI-VISA is version 3.4 and currently using labview 8.0
Have you tried reading through this Getting Started Guide? http://zone.ni.com/devzone/cda/tut/p/id/4644
Also, could you post a screenshot of your front panel showing all your settings you are using for the configuration of the device?
I HAVE THE SIMILAR PROBLEM. I AM USING AGILENT 34970A DATA ACQUISITION SWITCH UNIT WITH SCANNING CARD AGILENT 34901A. I WANT TO MEASURE VOLTAGE MEASUREMENTS FROM THREE-CHANNEL PHOTODIODE BY USING CHANNELS 102, 103 AND 104 SIMULTANEOUSLY. I USED THREE TYPES OF THE SAME LABVIEW CODE GIVEN IN FIGURE 3 (http://zone.ni.com/devzone/cda/tut/p/id/4644) FOR READING THE VOLTAGE FROM EACH CHANNEL. HOVEWER THE OBTAINED VOLTAGE RESULTS ARE INCORRECT BY COMPARING WITH MONITORING RESULTS. FOR EXAMPLE, WHEN THE VOLTAGE AT THE CHANNEL 102 IS 2.12 V, THE VOLTAGE READOUT OBTAINED VIA THE SOFTWARE CHANGE RANDOMLY BETWEEN 0.6 V AND 11 V. THE SAME SITUATIONS ARE OBSERVED FOR THE OTHER CHANNELS, 103 AND 104.
CHOULD YOU PLEASE HELP ME TO SOLVE THIS PROBLEM?
You obviously have a broken keyboard so fix that first.
Then, run the examples that come with the driver. You also have the EZ Voltage function on your palette. Run that with a single channel and see what you get.
p.s. You certainly cannot get data 'simultaneously' so I don't understand that statement at all.
When I even use single channel i.e. 102, I am encountering the problem during the use of EZ Voltage.vi driver. I think that the spike voltage enter in internal DMM when the relay is opened in the relevant channel. When using the Display.vi, I haven't faced any problem. However, when using the Display.vi, there is no option of monitoring the channels 102, 103 and 104 conscutively in Lab view. Additionally, when using the "BenchLink Data Logger" to monitor the single channel like 104, there is no problem but instead of using of single channel when scanning mode is run with "BenchLink Data Logger" I faced the same spike based problems for all three channels selected. I fixed my keyboard by pressing "Caps Lock":-)