Aerospace, Defense, & Government
Academic & Research
Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
You can request repair, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Hey guys i am trying to get get output from the 9265 with chasis NI-cDAQ- 9174 but unable to do so.
I wish to have continuous current in range of 10-13mA.
1) I am able to successfully run the self test but when it comes to "test panel" in NI MAX, It starts but there is no current output at the channels .
2) When i am making a program in LabVIEW for generation of continuous current. I am getting "Error -200294 occurred at DAQmx Start Task.vi:7220002"
Please help me !! In solving my problem .. My attachment includes my labVIEW block diagram and picture of DAQ assistant code .
What do you need our team of experts to assist you with?
We'll be in touch soon!