Benchtop Measurement and Test
Distributed Measurement and Control
Systems Engineering Software
Perspectives showcases how NI sees what’s next in the world of test and technology.
You can request repair, RMA, schedule calibration, or get technical support. A valid service agreement may be required.
Provides support for NI data acquisition and signal conditioning devices.
Provides support for Ethernet, GPIB, serial, USB, and other types of instruments.
Provides support for NI GPIB controllers and NI embedded controllers with GPIB ports.
Just in case anyone has this problem:
There is a bug in the latest Keysight B2900 drivers. Specifically (but not necessarily limited to as I haven't been through all the code) the 'Configure Source.vi'
The ':RANG' command following after the ';AUTO:LLIM' command is out of sequence and results in an error 113 "Undefined header" being thrown due to bad syntax.
This ':RANG' command should actually follow the output level trigger command (as it was in the previous versions of the driver).
What do you need our team of experts to assist you with?
We'll be in touch soon!