I am using tektronix TDS6124C oscilloscope to monitor the signal at ethernet port
I am unable to configure the Sampling rate and no.of samples for the scope
Kindly can anyone send me an example for configuring the sampling rate.
Thanks for your reply
Yes i am using the drivers and i am unable to configure.
I am using the examples given with the drivers and if i change the time base in the code it is not reflecting in the scope
If you have any examples kindly send me if possible
I have a TDS 5104 at my desk and ran the "Acquire Multiple Waveforms Example" and changing the timebase worked correctly. Without changing the default value, the scope's screen showed 800 ns/pt, and when I changed the timebase from 0.005 to 0.05, the scopes screen displayed 8.0 us/pt.
Please try that example and only change the timebase the same way that I did and tell us how many s/pt the scope's screen displays.